TI and its respective suppliers and providers of content make no representations about the suitability of these materials for any purpose and disclaim all warranties and conditions with regard to these materials, including but not limited to all implied warranties and conditions of merchantability, fitness for a particular purpose, title and non-infringement of any third party intellectual property right. Enter the information you’ve collected about your device into the four fields shown in Figure 5. No license, either express or implied, by estoppel or otherwise, is granted by TI. Double-click on your device to open a VISA session to the device. Before you can communicate with your device using these functions, you need to set up the communication protocol using the VISA USB attributes.
|Date Added:||25 November 2005|
|File Size:||46.36 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The plugin software contained ni-gisa usb driver. On the second screen, select Let me pick from a list of drivers on my computer. Just thought I’d point that ni-visa usb device in case someone’s using Windows 7; so yes, i agree with everyone else about an update.
Using NI-VISA to Communicate with Your USB Device
Ni-visa usb device we have seen that the device doesn’t reset properly to complete the device installation and therefore you would ni-visa usb device the error that you see. I expect this is because the installation of ADCPro does not install the drivers.
On the CD that came cevice the kit is 5 files. All content and materials on this site are provided “as is”.
Is there a function that would list commands or a set of standard ni-visa usb device that would be a start. If you have further questions related to this thread, sub may click “Ask a related question” below. Mr Carlson – When you are finished the prompts, if you reset the HW is it recognized?
Configuring NI-VISA to Control Your USB Device – NI-VISA Help – National Instruments
Is this considered successful or I need to install it again? Installation instructions are included in ni-visa usb device header at the top of the INF file. You changed your response to Not Helpful. My troubles were from a lack of instruction in the users guide.
Also, it only worked when I uninstalled the weiusb driver that ni-visa usb device with the product CD consistent with article. Use of the information on this site may require a license from a third party, or a license from TI.
USB Instrument Control Tutorial
Each type of pipe transfers a different type of information. Ni-visa usb device Device Manager ni–visa open, expand the tree category appropriate to your device, ie: A value of —1 indicates a USB device does not support that type of pipe.
Worked great on a Mac Go to Support Forums. It is platform independent, bus ni-visa usb device, and environment independent. To read a property, select the View Attributes tab.
To test communication with this device, open MAX. A window similar to that shown in Figure 2 should open. The Installation Options window is displayed ni-visa usb device shown in Figure 7. Check with your device manufacturer n-visa your device command set.
In reply to Greg Hupp:. I found that the device wasn’t showing up after installing the INF file as explained.
Connecting Instruments via USB – National Instruments
uusb Make sure your device is selected in ni-visa usb device window shown in Figure 8, then click Next. When you have completed your configuration and verified communication with the instrument, you need a ni-visa usb device way to progress from interactive mode to programming mode so you can immediately begin writing your tests without a tedious transition process.
The document is good, but need to be update; I am able to put online one update for this document. June Part Number: I choose the adcpro-setup-withlvrt Content on this site may contain or be subject to nni-visa guidelines or limitations on use.